Message 4 of 7. So i can conclud that my chip actullay connect ok with labview. Also, if possible please let me know if I can find some elaborated material. I disable the outputs, and then query the max voltage level. All this works and retunrs good data.
|Date Added:||8 March 2012|
|File Size:||37.57 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Agilent Technologies / Keysight Technologies 8563EC
So i can conclud that my chip actullay connect ok with labview. Hello I realized a program for data acquisition with labview and agilentacquisition in real lanview works very well with a time you sampling from a second I have two different modules in the chassis.
Message 6 of 7. What kinds of instrument drivers support the EA?
Where can I download VXI I did most coding and have few questions about adding new steps in Profile. AUTO” be used for? I 8563ecc using Query TraceArray.
I am using Query TraceArray. Also, I know some quesry vi are their only, where I found queryfrequency. Here as shown in below screenshot, I am getting the amplitude with respect to data points upto It seems, when the program trace the data it only trace amplitude. I have tried that. Programmable Current Sinking In general, current sinking is a fixed value at some percentage of the full scale current output.
Why have you decided to not even look at the code on the existing driver. I have the right ports open and Wireshark shows that the communications are passing, but the values that I’m trying to communicate are not.
技術支援: EC 可攜式頻譜分析儀，9 kHz到 GHz [已過時] | Keysight (formerly Agilent’s Electronic Measurement)
If I need to look for some code, please let me know how to do it! Why did you decide not to use the existing driver? Message 1 of 7. No, all the B sample programs can only be used in the HP Basic for Windows operating environment. I am new to this group.
Agilent X Series Analyzer – IEEE (GPIB) Driver for LabVIEW – National Instruments
8563ev PX statistics example displaying 0. Contact us about this article. I have not a single clue how to get with learning labview each step by step. I have tried passing 0V, 20V, and the output from the max voltage back into its lahview input. Is it the problem that the command type sent from labview is string so that my chip cannot recognize that?
It is caused by a problem on the cash file of proxy server or on the browser of your PC.
I have tried with both the IVI function as well as the driver specific function. How to assign step numbers in Watlow Profile?
Or some kind of other method? Yes, using SCPI remote programming commands, follow these steps Measurement and 853ec Explorer gets different Visa resource names for serial ports each time I run it.
Keysight Technologies / Agilent Technologies hp856xe Analyzer
There is a way to read f0 and df that would be documented in the manual and implemented in the driver. Message 7 of 7. The problem is that when I query, it looks like my device is properly recognized and no Error, lavbiew no data from test instruments come.