Set and Read the voltage. Did you switch it to RS as directed on p. You can install the driver by double-clicking on the file. I got my drivers from the NI website. The mark LabWindows is used under a license from Microsoft Corporation. The supported model numbers are:
|Date Added:||28 May 2018|
|File Size:||43.2 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
E6331a browsing this forum: This tool automatically detects USB hardware connected to your host PC and adds a graphical representation to the window. With compact and modular form factors, NI’s PXI programmable power supplies and precision sources simplify the task of designing automated test systems by saving expensive rack space and reducing the need for multiple instrument form factors within a test system.
The attached vi shows proper use of error wires. By keeping the File Path input to the function blank, the program prompts the user to browse to a location on disk and give the file a name.
Labvies got myself a serial-usb-adapter and installed the drivers for it. Windows is a registered trademark of Microsoft Corporation in the United States and other countries. Set and Read the voltage. Here is the way it is done with my drivers.
LabVIEW Portal • View topic – Agilent EA issues
However, I have made the mistake by connecting just with a gender changer. Rename it Tracking Channel. I see troubleshooting by elimination sequence as follows: Or do I have to change e363a ea labview settings of the device itself?
It contains “Getting Started” information on using the driver in a variety of programming environments as well as documentation on IVI and driver specific methods and properties. These are necessary in case kabview error occurs.
Back to Top 2. Back to Top 4. The device is connected and it is switched on. Navigate to the ‘Hardware Interfaces’ palette to find the ‘Configure Tracking’ function. To help accelerate software development, all instrument drivers follow a consistent programming flow of: Da i can avoid this jumping between read and write.
NI develops and supports thousands of instrument drivers for third-party instruments to help you automate measurements. Create a control on this Read Output. You can also use the VISA resource wire.
I am a bit embarassed again.
AGILENT E3631A LABVIEW DRIVER DOWNLOAD
Back to Top 7. Is the driver for usb-to-rs properly installed device manager 2.
Select the PC from the SystemDesigner page. Navigate to the panel, and move the indicators from the Unplaced items onto the front panel. Delete the wires connecting Configure Current Limit. This can happen with your code because you have the setting of the outputs in parallel.
Message 10 of The procedure for detecting your device depends on the bus being used. Back to Top 6. You can also troubleshoot and make manual function calls to your instrument to ensure that it is working as expected before moving into automation.